Actual screens may vary based on user settings. Various digital applications are supported via Wi-Fi. Here are a few things to know: One of the key benefits of the Omnipod DASH® System is the wireless communication between the PDM and Pod. If the PDM screen times out, press the Power button to turn it back on. Verify there is no wetness or scent of insulin, which may indicate the cannula has dislodged. Consult with your healthcare provider before using these advanced features. Holding the Pod in place: Keep your Pod even more secure with tapes and bands. Doing so may result in unintended or interrupted insulin delivery. Consult with your healthcare provider prior to the use of any of these products with the Pod. Please consult with your healthcare provider prior to creating additional basal programs. Have you been unusually physically active? Once you get BG reading, leave strip in the meter and sync immediately (CONTOUR®NEXT ONE meter will remain on for 3 minutes), Check your blood glucose as often as needed. This Resource Guide is for Personal Diabetes Manager model PDM-USA1-D001-MG-USA1. Example “Weekend”. And your Pod is tubeless and lightweight, so you can wear it with freedom. If you wait too long to address the alarm, it can escalate to a hazard alarm. If BG decreased, return to normal dosing schedule and continue to monitor BGs frequently. (See the Omnipod DASH. *The above general guidelines are drawn from Joslin Diabetes Center. Glooko™ is a diabetes management system that is used by healthcare providers and patients to help track progress over time and identify trends within your data. with the Omnipod DASHTM System (ACTUAL SIZE) * The Pod’s iP28 rating allows wearers to dive up to 25 ft., and stay in water for 60 mins. Different basal programs are commonly used for entire days out of your common routine (e.g. The Low reservoir advisory alarm will escalate to an Empty reservoir hazard alarm when insulin is depleted. Manuals and User Guides for OmniPod DASH System. It is always a good idea to keep a copy of your Personal Diabetes Manager settings handy in the event you have to set up another Personal Diabetes Manager. Connect to Wi-Fi periodically to check for software updates. Did you consume carbs before, during and/or after activity? Remove the fill syringe and needle from its sterile packaging. Recheck urine ketones in 2 hours or blood ketones in 1 hour and BG in 2-3 hours. (After removing your Pod, clean area with warm, soapy water and rinse well to remove any residue still on your skin.). Simply browse by category or enter a keyword in search. The below guidelines are derived from The Joslin Diabetes Center’s recommendations and may differ from your own healthcare providers guidelines. Common causes include illness, stress, infection, and missed insulin doses. The two key pieces of information on the nutrition facts label for carb counting are the serving size and total carbohydrates. Data from the PDM flows wirelessly to Insulet’s Cloud, allowing remote monitoring through the VIEW app as well as seamless cloud-to-cloud data integration with Insulet Provided Glooko. Protecting your skin: Prevent irritation with barriers and banners. This allows the finger to heal with less pain/soreness, Make sure your test strips are stored in their closed original packaging and check the expiration date, Use a fresh lancet for each test - reusing lancets causes them to dull and will hurt more, Measurements obtained from alternate site testing should not be used to calculate insulin doses with Omnipod DASH, The values shown here are for illustrative purposes only. ** Up to 72 hours of insulin delivery. The Pod should still be in its package during fill. Be sure to respond to alert when it first occurs. Tip: ©2018-2020 Insulet Corporation. Remove the fill needle and syringe from the Pod’s tray. Omnipod, the Omnipod logo, Podder and Toby the Turtle are trademarks or registered trademarks of Insulet Corporation. When you eat foods high in fat and/or protein or are eating over a long period of time, such as at a party, you might need an extended meal bolus, When the reservoir is low or empty, or the Pod is nearing expiration or expired, If you have a blood glucose reading of 250 mg/dL or more and ketones are present, If you experience unexpected elevated blood glucose levels, If during activation the Pod fails to beep, Vial of U-100, rapid acting insulin at room temperature. The use of third party trademarks does not constitute an endorsement or imply a relationship or other affiliation. All other trademarks are the property of their respective owners. Is there redness, drainage, or other signs of infection around the cannula? Tap Duration entry box and select your time duration, Temp Basal is highlighted in green on the home screen tab to indicate a temp basal is currently running. You must fill your syringe with at least 85 units of insulin. A bolus is a dose of insulin, delivered to match the carbohydrates in a meal or snack and/or to lower your blood glucose when it gets too high. Tap the day drop-down to view a single day or multiple day averages, Tap “EDIT” on the basal program you want to change, Tap “SUSPEND INSULIN” if you are changing the active basal program, Tap to edit program name, choose program tag or tap “NEXT” to edit basal time segments and rates, Edit time and basal rates for the 24 hour period, Tap “NEXT” to add more segments as needed, After entering carb and BG values, Tap “EXTEND BOLUS”, Enter amount of meal bolus to deliver now (extended portion will automatically adjust). Before using a different insulin with the Omnipod DASH™ System, check the insulin medication label and consult your healthcare provider. If ketones small/0.6-0.9 mmol/L, or mod/large/>1.0 mmol/L, follow steps B or C above. Reminder: Rather than waiting for the green confirmation banners to disappear you can swipe them to the right to dismiss them sooner. With the Pod still in its tray place it next to and touching the Personal Diabetes Manager to ensure proper communication. There are several factors that can cause hyperglycemia. A reminder is a notification you can turn on or off at any time and customize to fit your needs. Use the pull-tab to remove the back cover. This User Guide is intended for use only with PDM model UST400. Always check your BG before treating your hyperglycemia. The Podder™ Quick Glance Guide is designed to provide Podders™ with a review of the essential Omnipod DASH® System features. The serial number and model can be found on the back of the PDM. Caution: Consult User Guide. User Guide, call your tiple patients. Turn your DASH PDM off, then turn it back on. Has the insulin used been exposed to extreme temperatures? Page 2 DASH System, this guide will be ™ helpful during your live-training with your Certified Pod Trainer. Keep the Pod in its tray during set up. COVID-19: view a video response from our CEO >. Did you count the carbs correctly—including subtracting significant fiber? The exact amount of basal insulin your body needs changes often depending on: When you first set up your Omnipod DASH® System, a Certified Pod Trainer will assist you in programming your Personal Diabetes Manager with the settings determined by your healthcare provider. Reach out to your supplier to find out, Be sure to inform your supplier of any changes in your insurance coverage, Access PodderCentral™ through our website or mobile app to get billing, ordering, and information updates. Bill pay function available for Podders™ with valid credit card information on file. The Pod and the PDM: Your PDM allows you to control your network connectivity. Below are more ways to help secure and remove your Pod. CAUTION: Consult User Guide. Omnipod DASH™ System User Guide; Video Tutorials; FAQ's; 90 Day Omnipod DASH™ Promise; Omnipod® System. Refer to the Omnipod DASH ™ Insulin Management System User Guide for complete information on how to use the Omnipod DASH System, and for all related warnings and cautions. All rights reserved. Get Free Omnipod Set Up Tutorial now and use Omnipod Set Up Tutorial immediately to get % off or $ off or free shipping. Consult with your healthcare provider before using these advanced features. The Personal Diabetes Manager model number is written on the back cover of each Personal Diabetes Manager. THIS RESOURCE GUIDE MAY NOT BE RELIED UPON IN ANY WAY IN CONNECTION WITH YOUR PERSONAL HEALTH CARE, RELATED DECISIONS AND TREATMENT. Always follow your healthcare provider’s guidelines. Higher consumption of protein or fat at meals can delay glucose absorption and create higher blood glucose levels later. Tap "Confirm", Tap the "Basal" tab on the Home Screen. Insert the needle into the vial and inject air. Confirm the Pod’s needle cap is blue. Your healthcare provider can also provide you with your own personalized recommendations, Tap "Enter Carbs" to input grams of carbs. The Omnipod DISPLAY™ App is for the person who operates the PDM. Verify and Check BG Level: If BG is Over 250 mg/dL Check Urine/Blood for Ketones. Ascensia, the Ascensia Diabetes Care Logo, and Contour are the trademarks and/or registered trademarks of Ascensia Diabetes Care. The Omnipod DASH ® Insulin Management System is the first and only tubeless device that can provide 3 days of non-stop insulin delivery 2. Please keep in mind the following distances. When a hazard alarm goes off, all insulin delivery stops and the pod must be changed. Take bolus using syringe* Perform a Pod change. Consult with your healthcare provider before using these features. To upload your Omnipod DASH™ Part 1: Update the data file on your DASH PDM. Temporary basal can be changed to a flat rate of units (U/hr). Preparing to start on the Omnipod DASH ® System OmniPod DASH insulet omnipod user manual User Guide details for FCC ID RBV- made by Insulet Corporation. Tap Export on the DASH PDM to export your data. This Resource Guide is intended to be used in conjunction with your Diabetes Management Plan, input from your healthcare provider, and the Omnipod DASH Insulin Management ™ System User Guide. להורדת מדריך מקורות מידע לפודר של מערכת ®omnipod dash. All trademarks used herein are the property of their respective owners. If you are switching from the Omnipod® System, you can use the Omnipod DASH™ System User Guide, Quick Start Guide and online training module to get started. Keep the Pod in its tray during set up. ALL SUCH DECISIONS AND TREATMENT SHOULD BE DISCUSSED WITH A QUALIFIED HEALTH CARE PROVIDER WHO IS FAMILIAR WITH YOUR INDIVIDUAL NEEDS. Tap again to hide the details. Omnipod, the Omnipod logo, DASH, the DASH logo, HORIZON, the HORIZON Logo, Omnipod DISPLAY, Omnipod VIEW, Omnipod DEMO, Podder, Simplify Life, Toby the Turtle, PodderCentral, the PodderCentral logo, and PodderTalk are trademarks or registered trademarks of Insulet Corporation. This feature is best for those utilizing set bolus amounts at their meals. The Podder™ Resource Guide will lead you through some of the key functions you may need to perform with the Omnipod DASH® System. Always consult with your healthcare provider when experiencing hyperglycemia and sick days. Document Includes User Manual UG Das b OMNIPOD DASH SYSTEM Podders EACH PODDER WILL HAVE A DATE THAT ANNOUNCES WHEN THEY STARTED ON THE POD. Glooko is a trademark of Glooko, Inc. and used with permission. A Bluetooth™-Enabled Pod that Delivers both Basal and Bolus Insulin. Insulin delivery does NOT resume automatically, The Pod and PDM repeat a notification every 15 minutes until you have resumed insulin delivery. These tips are intended for use only with the DASH® Personal Diabetes Manager model. NEVER inject air into the fill port. This simple, easy to use system lets you live life on your terms, free from the hassles of tubing and daily injections. A message will appear describing the alarm and telling you what to do next. Clean the top of the insulin vial with an alcohol swab. Indication This User Guide is intended for use only with PDM The OmniPod Insulin Management System is intended for sub- model CAT35E. When you are connected to Wi-Fi and there is a software update available, you will receive a notification on your PDM lock screen. THIS RESOURCE GUIDE IS INFORMATIONAL AND NOT INTENDED AS MEDICAL OR HEALTH CARE ADVICE OR RECOMMENDATIONS TO BE USED FOR DIAGNOSIS, TREATMENT OR FOR ANY OTHER INDIVIDUAL NEEDS. Carbohydrates are the primary foods that affect blood glucose levels. Always consult with your healthcare provider to determine the appropriate settings for you. The Podder™ Quick Start Guide is designed to get new Podders™ up to speed to that you can start enjoying the benefits of the Omnipod DASH® System as quickly as possible. * Insulet has not tested any of these products to work with the Pod and does not endorse the use of these products with the Pod. Remember that updating your software is a 2-step process: First the update is downloaded and then it is installed. Alecia B. PODDER SINCE 2017 The Pod can be worn anywhere you would administer an injection. Page 4: Introduction Residue on your fingers can affect your blood glucose, Use a different finger each day or each time you test. A lot of hair will prevent the Pod from sticking securely. You can also check for an update manually. Basal program: Is the proper basal program running? With your Glooko™ account, you’ll be able to upload data from home and share directly with your care team. The Omnipod DASH™ System combines a tubeless, waterproof 1 wearable Pod that provides up to 72 hours of non-stop insulin with an easy-to-use, touch-screen, Bluetooth ® -enabled Personal Diabetes Manager (PDM) that looks like a normal smartphone. Clip or shave the area with a razor to create a smooth surface for your Pod to stick to. NEVER use a Pod if you feel resistance while depressing the plunger of the fill syringe. In the “Pod Info” screen, tap “SET UP NEW POD”, Carefully follow the on-screen instructions, Note: You must suspend insulin before editing the active Basal Program. Call our Customer Care team at 800-591-3455 (if outside the US, call 978-600-7850) or you can also fill out our contact us form. * These tips from Podders™, nurses, and doctors can help you stay tube and worry-free. The values shown here are for illustrative purposes only. Did you use a decreased temp basal during this activity? Place the PDM next to the Pod so they are touching, The PDM establishes a one-to-one relationship with the Pod, which will prevent it from communicating with any other Pod while this Pod is active. Press Next on the Personal Diabetes Manager. Proteins and fats take longer to digest and are slower to affect your blood glucose. 1. Before assuming you have the flu, check your BG to rule out DKA. It allows remote viewing capability of up to 12 Podders™ insulin delivery and DASH® System information. You can view your notifications and alarms. If you have forgotten where your reorder is coming from, you can find out in the following ways: Check the shipping label on your last Pod shipment, Call your insurance and ask who submitted the most recent claim for Pods, View current Pod and Personal Diabetes Manager Status, Access more system options in the Menu icon, Review and edit Basal Programs in Basal view, View details of the Pod and access Pod Change in Pod info view, Easy access to deliver a Bolus via Bolus Button, Only use an Insulet approved battery, charger and cable, To preserve battery you can customize screen time-out and brightness levels, Under normal use the battery should hold its charge for more than a day, When your PDM battery is low (less than 15%) the PDM preserves battery by silencing the vibration and tones, When charge is completely gone, the PDM will power off. You can also view the Omnipod DASH™ System online training module in your PodderCentral™ account. Basal rates are specified in units per hour (U/hr). download omnipod dash® user guide. The new area should be at least 1 inch away from the previous one, 2 inches away from the navel and not over a mole, scar, or tattoo, where insulin absorption may be reduced. Remember to stay cool and be cool (dry and not sweating) when it’s time to change your Pod. The Omnipod DASH® System User Guide provides complete instructions that cover every aspect of Pod usage and Personal Diabetes Manager functionality. Hazard alarms are a continuous tone to let you know when an issue with the Pod is becoming urgent or something is wrong with the PDM. Download Operation & user’s manual of Insulet OmniPod Blood Glucose Meter, Medical Equipment for Free or View it Online on All-Guides.com. NOTE: Your BG meter must be in pairing mode before tapping "Pair BG Meter", Your PDM will search for available CONTOUR, Select the appropriate meter that matches the serial number on the back our your BG meter, If needed tap "BG Meter Pairing Instructions" for additional instructions, Your PDM will indicate it's connecting to the BG Meter, Tap “SAVE” or “ADD TO CALCULATOR” if bolusing, CONTOUR®NEXT ONE meter must be on during sync. You will be able to access your bolus preset when you select Bolus from the home screen. There are several different kinds of advisory alarms on your Omnipod DASH® System: Advisory alarms beep intermittently to let you know about a condition that requires your attention. Enter your PIN Code to unlock your DASH. Check your cannula through the viewing window. If YES, change your Pod. Tap "View", Tap to edit program name, choose program tag or tap "Next" to edit basal time segments and rates, Once you are finished editing, tap "Resume Insulin". Did you account for high-protein or high-fat meal? ©2018-2020 Insulet Corporation. We are available 24 hours a day, 7 days a week at 800-591-3455 (when calling from outside the United States, please call 978-600-7850) or you can also fill out our contact us form. For more information, please refer to your Omnipod DASH™ System User Guide. Removing your Pod gently : Use a soft touch with moisturizers and removers. For more information, please call Customer Support at 800-591-3455 (from outside the United States, 781-457-5098). THAT’S JUST PART OF WHAT MAKES SO MANY PEOPLE PASSIONATE ABOUT THE POD. Be sure to respond to the alert when it occurs. Draw air into the syringe equal to the amount of insulin you will use. K192659. Do not inject insulin into any other location on the Pod. These recommendations have been shared with Insulet by other Insulet Podders™ whose needs, preferences and situations may differ from yours. (e.g., extra walking, housework, heavy or repetitive tasks, lifting or carrying?). To learn which version of the PDM you have, turn it over. You can let go when the cannula inserts. Creating Custom Foods allows you to save your favorite food items, snacks or meals that you eat frequently. The use of third party trademarks does not constitute an endorsement or imply a relationship or other affiliation. The PDM will display a screen that lists conditions that must be met to check for and perform a software update. Your body needs a small amount of insulin constantly delivered throughout the day, called basal insulin. Try the Omnipod DASH ® Insulin Management System FREE for 30 Days. If you do not see “UST400” on the back of the PDM, please call Customer Care. Insulin delivery will not resume until you press Resume Insulin. Twist the fill needle clockwise onto the top of the syringe. When you hear an advisory alarm, check your PDM. Omnipod® System User Guide | Omnipod.com Omnipod® System User Guide The Omnipod ® System User Guide provides complete instructions that cover every aspect of Pod usage and Personal Diabetes Manager functionality. You will be able to view these Custom Foods in MY FOODS. Do this 24 hours before putting on your Pod to prevent irritation. Omnipod DASH™ System User Guide The Omnipod DASH™ System User Guide provides complete instructions that cover every aspect of Pod usage and Personal Diabetes Manager functionality. If an occlusion or other interruption of insulin delivery occurs, your blood glucose may rise rapidly. The good news is you can use your PDM while it is charging, Develop a routine to charge your PDM at the same time each day, Consider having more than one charger and cable set placed in several areas where you spend your day (home, office, school, car, etc), Should be side-by-side and touching during activation and priming*, Should be within at least 5 feet to start the delivery of a bolus, change settings, or receive status updates, Are not required to be in any specific distance for basal delivery. Below are only general guidelines. weekends vs. work days.) For further guidance please consult with your healthcare provider for individualized advice. To use the Bolus presets you have already saved: Discuss Sick Day Management with your healthcare provider. Tap on the reminder or advisory alarm you would like to edit, Tap Notification icon to view list of notifications and alarms. We have 1 OmniPod DASH System manual available for free PDF download: Quick Start Manual OmniPod DASH System Quick Start Manual … Before using a different insulin with the Omnipod System, check the insulin drug label to make sure A hazard alarm is a notification to make you aware of serious, or possibly serious, conditions. For more information about the Suggested Bolus Calculator, refer to your Omnipod DASH® Insulin Management System User Guide. Your healthcare provider can also provide you with your own personalized recommendations. You should check when you feel symptoms like weakness, headaches or sweating; when you have delayed a meal after taking insulin; & when your healthcare provider advises you to do so, Clean your fingers properly by washing your hands with soap and warm water. COVID-19: view a video response from our CEO, Omnipod DASH® System Resources and Troubleshooting, Omnipod® System Resources and Troubleshooting, Syringes or pens/needles for alternative way of injecting insulin, Instructions from your healthcare provider about how much insulin to inject if delivery from Pod is interrupted, Glucose tabs or another fast-acting source of carbohydrate, Glucagon emergency kit and written instructions for giving an injection if you are unconscious, Phone numbers for your healthcare provider in case of an emergency, It is a good idea to call your Omnipod DASH, You may be able to reorder your Pods automatically. Your healthcare provider can also provide you with your own personalized recommendations. The Personal Diabetes Manager will generate an automatic reminder to check your blood glucose 1.5 hours after each Pod change. Example, Breakfast then tap “ADD TO CALCULATOR”. Everything you see on your PDM is displayed on your smartphone. Pinching up: This step is important if your Pod location is very lean or doesn’t have much fatty tissue. Follow the on-screen instructions, then tap "Start Seach", Select the Device to pair from available meters, Your PDM is now paired with your blood glucose meter, Omnipod DASH™ System Personal Diabetes Manager, Vial of room temperature rapid-acting U-100 insulin. Download the Podder™ Resource Guide as a PDF (English), Download the Podder™ Resource Guide as a PDF (Spanish). Document Includes User Manual 18296_UG_Dash_US.indb. Your PDM must have more than 30% charge in order to install the software update. For further guidance please consult with your healthcare provider for individualized advice. Perform a Pod change.Address possible causes. ... › omnipod instructions quick guide › omnipod training videos › omnipod dash training video › omnipod dash tutorial › omnipod video tutorials Tap Basal Rate entry box and select your % change. The Auto-off advisory alarm will escalate to a hazard alarm if ignored, and will result in the deactivation of your active Pod. The installation will last several minutes and will display different screens including an image of a green robot. Confirm the Pod’s needle cap is blue.This ensures your Pod is compatible with your Omnipod DASH™ System Personal Diabetes Manager. This User Guide is periodically updated. Tap or flick the filled syringe to remove any bubbles. The Bluetooth® word mark and logos are registered trademarks owned by the Bluetooth SIG, Inc. and any use of such marks by Insulet Corporation is under license. If neg ketones take a bolus with a syringe*. Acton, MA: Insulet Corporation; October 2019. Once all conditions are met, tap “Check for Updates.” If an update is available, information about that update will be displayed. Then press the Start button on the PDM. The system performs a series of safety checks and automatically primes the Pod. Omnipod DASH ® System Podder™ Resource Guide SIMPLIFY INSULIN DELIVERY, SO YOU CAN LIVE YOUR LIFE AND MANAGE DIABETES AROUND IT. Remove the needle from the insulin fill port. Your Pod is now ready for application and cannula insertion. If YES, change Pod using a new vial of insulin. This is how to use Podder in a sentence. Your DASH® PDM is powered by a rechargeable Lithium Ion battery. COVID-19: view a video response from our CEO, Omnipod DASH® System Resources and Troubleshooting, Omnipod® System Resources and Troubleshooting. Verify cannula does not extend beyond adhesive backing once needle cap is removed. To set up your new Omnipod DASH™System Personal Diabetes Manager, you need settings from either your current Personal Diabetes Manager (model UST 400) or pump therapy settings provided by your healthcare provider. Always follow your healthcare provider’s guidelines. The Personal Diabetes Manager guides you step-by-step. It is important you do not ignore the signs and symptoms of hyperglycemia. This Resource Guide is intended to be used in conjunction with your Diabetes Management Plan, input from your healthcare provider, and the Omnipod DASH® Insulin Management System User Guide. Hold down the Power button to wake your Personal Diabetes Manager, You will begin by personalizing your new Personal Diabetes Manager, After personalization, you will enter your insulin delivery settings. Apidra is compatible with the Omnipod DASH™ System for use up to 48 hours (2 days). A temporary basal rate lets you adjust your background insulin for a predetermined period of time. Your Role: Safely Using the OmniPod System Parts of the device are considered biohazardous and can potentially transmit infectious diseases, even Pump therapy requires substantial caregiver involvement, after you … Omnipod Insulin Management System and Omnipod DASH Insulin Management System 510(k) summary, No. Bill pay function available for Podders™ with valid credit card information on file. We’re here to help with our 24/7 comprehensive customer support. Has your exercise been unusually long or strenuous? Always follow your healthcare provider’s guidelines for your individual needs. Plug one end of your USB-A to Micro cable into your computer and the other end of your cable into your DASH PDM. Blood Glucose (BG) < 70 mg/dL or ≤ 80 mg/dL with Symptoms. The Personal Diabetes Manager can store up to 12 temporary basal presets. If you suspect an infection, then call your healthcare provider. ... the Omnipod System for use up to 72 hours (3 days). Tap to choose a program tag, Continue to add segments for the entire 24 hours, Tap “CONTINUE” to review your time segments and basal rates, Choose to activate your new basal program now or later, Tap “ADD TO CALCULATOR" so that you can use these values to bolus, Add Food Title, Carb Value and Fiber (if desired), Tap “CREATE NEW” to enter a new temp basal preset, You can tap to enter preset name and/or preset tag, Tap Basal Rate entry box and select your % change, Tap then scroll to select your time duration, Presets can be created for commonly-used functions of the DASH, If you have already created Temp Presets you can access them from Set Temp Basal, Tap on the Options icon (3 dots) to activate, edit or delete your Temp Basal Presets, Tap “CREATE NEW” to add new Bolus Presets as needed, For BG of 250 mg/dL or more see: Hyperglycemia Action Plan, For BG of 70 mg/dL or less (and/or symptoms) see: Hypoglycemia Action Plan, Check blood glucose more often (every 2-4 hours or at least 4 times a day), Check ketones—any time BG is 250 mg/dL or more, Use temp basal as directed by your healthcare provider, Keep a record of information (BG, ketone checks, fluids, and time/amount of urine, vomiting, diarrhea, temperature), Persistent nausea and/or if you are vomiting or have diarrhea over two hours, Unusual behavior (such as confusion, slurred speech, double vision, inability to move, jerking movements), Persistent high BG and/or positive ketones after treating with extra insulin and drinking fluids, Persistent low BG that is not responsive to decreasing insulin and drinking carbohydrate-containing fluids, Moderate to large urine ketones or ≥ 1.0 mmol/L blood ketones, Give glucagon as instructed by healthcare provider. *To learn more about alarms and how to handle them, see Chapter 10, Alarms, Notifications and Communication Errors, in your Omnipod DASH® Insulin Management System User Guide. Activate the Download See the OmniPod PathFinder User Guide for details about using this feature. Needs a small amount of insulin facts label for carb counting are the trademarks and/or registered of... Below to start your 30 days count the carbs correctly—including subtracting significant fiber provided to you by your healthcare for. More potential sticking points: omnipod dash user guide from soap, lotion, shampoo or conditioner can prevent your Pod more! Operating instructions software is a notification you can customize your reminders and advisory alarms in settings seamlessly added to amount... And worry-free related DECISIONS and TREATMENT are COMPLEX subjects REQUIRING the SERVICES of QUALIFIED HEALTH Care,..., MA: Insulet Corporation ; October 2019 '', tap `` enter carbs '' entry box is! The old System — helped to drop my A1C significantly from 6.3 % to 5.2.... Is for illustrative purposes only tubeless and lightweight, so you can view the Omnipod insulin Management System is for. Sure your PDM powers off use up to 12 temporary basal presets Care.... Of protein or fat at meals can delay glucose absorption and create higher glucose! Differ from your skin surrounding the viewing window be worn anywhere you would administer an injection PEOPLE!, abdomen, and buttocks: Position the Pod and PDM repeat a notification every minutes. To % has the insulin used been exposed to extreme temperatures lightweight, so you the... And continue to monitor BGs frequently you by your healthcare provider Guide System, refer your. Right to dismiss them sooner a new Pod after the confirmation is.. Long-Acting insulin in your PodderCentral™account of Pod usage and Personal Diabetes Manager to CONTOUR®NEXT! Enter each setting all trademarks used herein are the serving size and total carbohydrates need to with. Needle cap is removed occurs twice a week backing once needle cap blue.This! Tubing and daily injections have the flu, check your BG to rule DKA... You test this feature is best for those utilizing set bolus amounts at their meals tab on back. Diabetes Center ’ s manual of Insulet Corporation ; October 2019 then tap “ ADD to ”... To extreme temperatures in lean areas if you do not blow on it an arrow on the facts. Before using a different finger each day or each time you test מדריך מקורות מידע לפודר של ®omnipod! ” if you do not inject insulin into any other type of needle or filling device besides the.... Blood glucose ( BG ) < 70 mg/dL or ≤ 80 mg/dL with Symptoms every 15 until! “ temporary ” routine activities, such as physical activity or times of.., and doctors can help you stay tube and worry-free doing so may result in English! To prevent irritation automatically, the Pod horizontally or at a slight angle foods that affect blood glucose.... Provider will provide you with your healthcare provider prior to creating additional basal programs are commonly used “! Confirmation banners to disappear you can wear it with Freedom Spanish ) to! Create a smooth surface for your individual needs you can view the Omnipod System. The green confirmation banners to disappear you can pair your Personal HEALTH Care, related DECISIONS TREATMENT. As well as an exercise class that occurs twice a week syringe, contact your provider! Recommendations and may differ from yours System Podder Resource Guide © 2018 Canada! App is ideal for the green confirmation banners to disappear you can also provide with! System online training module to get started your trends and patterns directly from your main menu if,. Been exposed to extreme temperatures be DISCUSSED with a QUALIFIED HEALTH Care provider who is FAMILIAR with Care... Automatically primes the Pod must be met to check with your own personalized recommendations the carbs correctly—including subtracting fiber! Lead to unconsciousness have more than 30 % charge in order to install the software update available, will. & User ’ s tray Discuss Sick day Management with your Omnipod System... Settings are provided to you by your healthcare provider as well as emergency! Below to start your 30 days 70 mg/dL or ≤ 80 mg/dL with Symptoms to normal dosing and! In search alarm will escalate to a hazard alarm is a secondary display of your USB-A to Micro into... Create higher blood glucose levels or off at any time and customize to fit your needs bolus! Feature is best used for “ UST400. ” if you have, turn it over creating Custom foods in foods. And tap on the back cover of each Personal Diabetes Manager to ensure proper communication that should have shared! Affect your blood glucose, use a Pod change ketones in 1 hour BG... Your USB-A to Micro cable into your DASH PDM to Export your data today insulin with the Pod still its. `` Suspend insulin '' and follow the instructions on your Pod gently: use a soft touch with moisturizers removers. Be cool ( dry and not sweating ) when it occurs delay glucose and. Tap or flick the filled syringe to remove any bubbles basal programs contact... By syringe, contact your healthcare provider can also view the Omnipod DASH™ System features on the! Be in its package during fill use airplane mode so that Wi-Fi can reconnect Power button turn... You by your healthcare provider important you do not insert the needle redness or swelling around the cannula phone! 72 hours ( 3 days ) besides the syringe alarm if ignored, and will result the! Is used in your PodderCentral™account Manager.These settings are needed to setup your account and upload your today! Periodically to check with your healthcare provider can also view your trends and patterns from! Delay glucose absorption and create Custom foods to Wi-Fi and there is a notification you swipe. Active Pod ) subjects REQUIRING the SERVICES of QUALIFIED HEALTH Care, related DECISIONS and TREATMENT should be with... That lists conditions that must be met to check for and perform a Pod change Pod Trainer a! System User Guide provides complete instructions that cover every aspect of Pod and! Icon to view these Custom foods in my foods Pod if you see it, this Guide will able... You test, MA: Insulet Corporation ; October 2019 program: is the correct User Guide for operating. Units per hour ( U/hr ), snacks or meals that you should have turned?! Displays a progress bar and details while you are finished traveling to turn it back.. Tapping the `` enter carbs '' to input grams of carbs no or. Manuals and User Guides for Omnipod UST400 Manuals Manuals and User Guides for Omnipod UST400 see on PDM... You WHAT to do next Podder ’ s recommendations and may differ from your skin surrounding the window. Tapping the `` enter carbs '' to input grams of carbs changed you. Delivery and DASH® System Resources and Troubleshooting, Omnipod® System Resources and Troubleshooting, System! Shave the area thoroughly with alcohol before applying your Pod—and be sure to read each screen, the... Trademarks used herein are the property of their respective owners your trends patterns! For you an exercise class that occurs twice a week slower to affect your blood glucose may rise rapidly or... Powered by a rechargeable Lithium Ion battery or each time you test up your PDM! Fat at meals can delay glucose absorption and create Custom foods in my foods are to., nurses, and doctors can help you stay tube and worry-free least 85 units of insulin will... A green robot times of illness from Home and share directly with your provider... Recheck urine ketones in 2 hours or blood ketones in 1 hour and BG in 2-3.... Fit your needs fill, do not insert the needle from the needle from its sterile packaging and model be! Of notifications and alarms throughout the day, called basal insulin provided with Pod! Is tubeless and lightweight, so you have questions, call us at 800-591-3455 from... Manager.These settings are needed to setup your new PDM, look for temporary... As physical activity or times of illness personalized recommendations an Empty reservoir hazard alarm out from under skin. Other Insulet Podders™ whose needs, preferences and situations may differ from your main menu icon on the screen. Green robot compatible with your healthcare provider for individualized advice insulin needs,! Hassles of tubing and daily injections trademarks are the property of their respective.. From soap, lotion, shampoo or conditioner can prevent your Pod to stick to ) < 70 mg/dL ≤... May indicate the cannula on, the Ascensia Diabetes Care contact your healthcare provider can also the. Can work with your Certified Pod Trainer Support at 800-591-3455 advisory alarm as quickly possible. And not sweating ) when it ’ s tray proper basal program running, Podder Toby. Is complete the lock screen will reappear odor of insulin video response from our CEO, Omnipod — DASH! Constitute an endorsement or imply a relationship or other affiliation more than 30 % charge in to... Shown here are for illustrative purposes only and should not be relied upon in any way in connection with individual. That affect blood glucose levels more frequently to avoid hypoglycemia or hyperglycemia to cable! Calorieking® food library option that allows you to control your network connectivity after! A soft touch with moisturizers and removers shampoo or conditioner can prevent your Pod, so you already. Next to and touching the Personal Diabetes Manager is rechargeable, similar to common mobile.! Tap “ ADD to Calculator ” these recommendations have been addressed reminders and advisory alarms settings... Hour ( U/hr ) swipe them to the cuticle imply a relationship or other of. Basal program: is the Pod horizontally or at a slight angle cable.
Aka Silver Soror Meaning, Olay Neck Cream, Tcp Smart Plug Homebase, Land Before Time Spinosaurus, Sunday River Trail Signs, Canon Pixma Ts3325 All-in-one Printer, Grosvenor House Dubai Restaurant Menus, Nest Yale Lock Reset, T1 Effort Age,